SWCNT-1: Single-Wall Carbon Nanotube Certified Reference Material
This Certified Reference Material is primarily intended for use in the validation of procedures and development of methods for the determination of trace metals in carbon nanotubes or those of a similar matrix. It has been gamma sterilized to support studies of environmental health and safety and nanotoxicology. SWCNTs have been identified as an engineered nanomaterial for which thorough characterization is needed for assessments of environmental, health and safety testing. Parameters of interest include elemental/molecular composition, specific surface area, surface chemistry, particle size and distribution, morphology/shape/form, surface charge and crystal structure, amongst other variables. SWCNT-1 serves as a stable and homogeneous representative test material to foster development and validation of international consensus-based standards to further rigorous physico-chemical characterization of such materials.
A unit of SWCNT-1 CRM consists of a bottle containing nominally 100 mg of SWCNT material under an argon atmosphere.
Table 1 summarizes those trace metals for which certified values have been established for this material. The expanded uncertainty (UCRM) in the certified value is equal to UCRM = kuc where uc is the combined standard uncertainty calculated according to the JCGM Guide and k is the coverage factor. It is intended that UCRM accounts for every aspect that reasonably contributes to the uncertainty of the measurement. A coverage factor of 2 was applied for all elements. The table below lists certified values for SWCNT-1, expressed on a dry mass basis.
|Co (a,b)||15.9 ± 1.0||g/kg|
|Ni (b,c)||14.4 ± 0.8||g/kg|
|Mo (b,c)||7.3 ± 1.1||g/kg|
|Fe (b,c)||2.2 ± 0.2||g/kg|
|Pb (c)||6.8 ± 0.9||mg/kg|
|Hg (c)||< 10 Table 1 note a||mg/kg|
- Table 1 note a
For the purposes of uncertainty propagation, this value (x) could be interpreted as x/2 ± x/(2√3) where the uncertainty value is derived assuming a rectangular (uniform) distribution within the interval 0 to x.
- a - inductively coupled plasma mass spectrometry; quantitation by method of additions
- b - instrumental neutron activation analysis
- c - isotope dilution inductively coupled plasma mass spectrometry
|Element||Mass Fraction Table 2 note a||Units|
|Al||494 ± 94||mg/kg|
|Au||0.139 ± 0.017||mg/kg|
|B||107.3 ± 3.1||mg/kg|
|Ca||2650 ± 300||mg/kg|
|Cr||285 ± 26||mg/kg|
|H||423 ± 0.990||mg/g|
|K||3220 ± 200||mg/kg|
|La||0.96 ± 0.27||mg/kg|
|Mn||135.8 ± 1.8||mg/kg|
|Na||167 ± 7||mg/kg|
|Ti||193 ± 22||mg/kg|
|V||4.37 ± 0.31||mg/kg|
- Table 2 note a
Uncertainty is an expanded uncertainty (k = 2) about the mean (n = 4) calculated by combining between-method and within-method variances following. A comprehensive description of the methodologies and comparison of individual results is available.
|Surface area Table 3 note a||Units|
|329 ± 6 Table 3 note b||m2/g|
|Value Table 4 note a||Units|
|Peak 1||502.9 ± 1.2||°C|
|Peak 2||602.7 ± 0.9||°C|
|Residual mass Table 4 note b||83.0 ± 5.8||mg/g|
- Table 4 note a
Expanded uncertainty calculated as U = k (standard deviation divided by square root of 21) for k = 2 based on replicate measurement cycles undertaken over a period of 4 months on fresh and "aged" samples.
- Table 4 note b
Dry mass basis.
|G/D Ration Table 5 note a|
|49.1 ± 4.8|
- Table 5 note a
Mean of peak heights of 13 determinations conducted over an 18 month stability study at NRC. Expanded uncertainty is calculated as U = k(standard deviation divided by square root of 13) for k = 2.
|Tube diameter Table 6 note a, nm|
|1.5 ± 0.2 Table 6 note b|
- Table 6 note a
Calculated following : ωRBM = 234/d + 10; TEM imaging provides 1.5 ± 0.2 nm (n = 54) and AFM imaging provides a height of 1.5 nm (n = 3).
- Table 6 note b
Expanded uncertainty calculated as U=k(standard deviation divided by square root of 17)
for k = 2 based on RBM frequencies in the range 125 – 225 cm-1 excited with 514.5, 632.8 and 785 nm sources.
Date of issue: June, 2013
Date of expiry: June, 2023
Revised: March 2016 (editorial update), March 2018 (date of expiry extended and Mg value removed from table 2)
Report a problem or mistake on this page
- Date modified: